HITACHI F-7000 Fluorescence Spectrometer
The device features:
Has high sensitivity, high scan speed, and a horizontal slit unique optical grating for high sensitivity.
Fluorescence trace analysis, materials used in the routine analysis in the field, pharmaceutical, and life sciences and high-tech fields such as quality control, and teaching. SNR 250:1 (PP), 800:1 (RMS), is conducive to the measurement of trace samples can be measured as low as 1 x 10-12mol / L fluorescein; machines carved concave diffraction grating for the F-2.2 series; scanning speed of 60,000 nm / min, to be completed within 1sec of the entire wavelength range of measurement; concentration range measurement is six orders of magnitude; phosphorescence lifetime can be carried out at room temperature for about 1ms analysis; automatic pre-scan function, optimization of the unknown sample measurement conditions; a proportion of energy monitoring to ensure the stability of photometric light measurement; horizontal slits designed to improve measurement sensitivity, reduce the amount of sample; slit high-resolution multi-level spectral resolution of 1nm; can automatically measure 96 sample wells on the microtiter plate; autosampler can be used, wavelength scanning can be performed, and three-dimensional scanning time spectroscopy; polarization can be used in combination with attachments to achieve fluorescence polarization and fluorescence polarization analysis.
The basic operating procedures
1. Start, preheat 15min.
2. Double-click the desktop icon corresponding open interface.
3. Data Processing Window Options content, including a total of three:
4. Close the lamps, then close the monitor windows.
MBBS in Wuhan University
Study Tips for Studying MBBS in China
Hebei North University Faculty Members Participated In National Model United Nations In Dc Washingto
Study MBBS in China for Saudi Arabian Students
What Is Studying Strategy Of Medical Students
Agriculture Science Ceremonies and Seminars